What is a galvanometer
A galvanometer is a device that is used to detect small electric current or measure its magnitude. The current and its intensity is usually indicated by a magnetic needle’s movement or that of a coil in a magnetic field that is an important part of a galvanometer.
Since its discovery in the 1800s, galvanometer has seen many iterations. Some of the different types of galvanometer include Tangent galvanometer, Astatic galvanometer, Mirror galvanometer and Ballistic galvanometer. However, today the main type of galvanometer type that is used widely is the D’Arsonval/Weston type or the moving coil type. A galvanometer is basically a historical name that has been given to a moving coil electric current detector.
What is a Moving Coil Galvanometer?
A moving coil galvanometer is an instrument which is used to measure electric currents. It is a sensitive electromagnetic device which can measure low currents even of the order of a few microamperes.
Moving-coil galvanometers are mainly divided into two types:
- Suspended coil galvanometer
- Pivoted-coil or Weston galvanometer
Moving Coil Galvanometer Principle
A current-carrying coil when placed in an external magnetic field experiences magnetic torque. The angle through which the coil is deflected due to the effect of the magnetic torque is proportional to the magnitude of current in the coil.
Moving Coil Galvanometer Construction And Diagram
The moving coil galvanometer is made up of a rectangular coil that has many turns and it is usually made of thinly insulated or fine copper wire that is wounded on a metallic frame. The coil is free to rotate about a fixed axis. A phosphor-bronze strip that is connected to a movable torsion head is used to suspend the coil in a uniform radial magnetic field.
Essential properties of the material used for suspension of the coil are conductivity and a low value of the torsional constant. A cylindrical soft iron core is symmetrically positioned inside the coil to improve the strength of the magnetic field and to make the field radial. The lower part of the coil is attached to a phosphor-bronze spring having a small number of turns. The other end of the spring is connected to binding screws.
The spring is used to produce a counter torque which balances the magnetic torque and hence helps in producing a steady angular deflection. A plane mirror which is attached to the suspension wire, along with a lamp and scale arrangement, is used to measure the deflection of the coil. Zero-point of the scale is at the centre.
Working of Moving Coil Galvanometer
Let a current I flow through the rectangular coil of n number of turns and a cross-sectional area A. When this coil is placed in a uniform radial magnetic field B, the coil experiences a torque τ.
Let us first consider a single turn ABCD of the rectangular coil having a length l and breadth b. This is suspended in a magnetic field of strength B such that the plane of the coil is parallel to the magnetic field. Since the sides AB and DC are parallel to the direction of the magnetic field, they do not experience any effective force due to the magnetic field. The sides AD and BC being perpendicular to the direction of field experience an effective force F given by F = BIl
Using Fleming’s left-hand rule we can determine that the forces on AD and BC are in opposite direction to each other. When equal and opposite forces F called couple acts on the coil, it produces a torque. This torque causes the coil to deflect.
We know that torque τ = force x perpendicular distance between the forces
τ = F × b
Substituting the value of F we already know,
Torque τ acting on single-loop ABCD of the coil = BIl × b
Where lx b is the area A of the coil,
Hence the torque acting on n turns of the coil is given by
τ = nIAB
The magnetic torque thus produced causes the coil to rotate, and the phosphor bronze strip twists. In turn, the spring S attached to the coil produces a counter torque or restoring torque kθ which results in a steady angular deflection.
Under equilibrium condition:
kθ = nIAB
Here k is called the torsional constant of the spring (restoring couple per unit twist). The deflection or twist θ is measured as the value indicated on a scale by a pointer which is connected to the suspension wire.
θ= ( nAB / k)I
Therefore θ ∝ I
The quantity nAB / k is a constant for a given galvanometer. Hence it is understood that the deflection that occurs the galvanometer is directly proportional to the current that flows through it.
Solved question: What is the purpose of introducing a cylindrical soft iron core inside the moving coil galvanometer?
Solution: The cylindrical soft iron core placed inside the galvanometer increases the magnetic field strength, thereby improving the sensitivity of the instrument. It also makes the magnetic field radial so that the angle between the plane of the coil and the magnetic lines of force remains zero at all times during the rotation of the coil.
Sensitivity Of Moving Coil Galvanometer
The general definition of the sensitivity experienced by a moving coil galvanometer is given as the ratio of change in deflection of the galvanometer to the change in current in the coil.
S = dθ/dI
The sensitivity of a galvanometer is higher if the instrument shows larger deflection for a small value of current. Sensitivity is of two types, namely current sensitivity and voltage sensitivity.
- Current Sensitivity
The deflection θ per unit current I is known as current sensitivity θ/I
θ/I = nAB/k
- Voltage Sensitivity
The deflection θ per unit voltage is known as Voltage sensitivity θ/V. Dividing both sides by V in the equation θ= (nAB / k)I;
θ/V= (nAB /V k)I = (nAB / k)(I/V) = (nAB /k)(1/R)
R stands for the effective resistance in the circuit.
It is worth noting that voltage sensitivity = Current sensitivity/ Resistance of the coil. Therefore, under the condition that R remains constant; voltage sensitivity ∝ Current sensitivity.
Figure of Merit of a Galvanometer
It is the ratio of the full-scale deflection current and the number of graduations on the scale of the instrument. It also the reciprocal of the current sensitivity of a galvanometer.
Factors Affecting Sensitivity Of A Galvanometer
a) Number of turns in the coil
b) Area of the coil
c) Magnetic field strength B
d) The magnitude of couple per unit twist k/nAB
Solved Question: Increase in current sensitivity results in an increase in voltage sensitivity of a moving coil galvanometer. Yes or no? Justify your answer.
Solution: No. An increase in current sensitivity of a moving coil galvanometer may not necessarily result in an increase in voltage sensitivity. As the number of turns (length of the coil) is increased to increase the current sensitivity of the device, the resistance of the coil changes. This is because the resistance of the coil is dependent on factors like the length and area of the coil.
Applications of Galvanometer
The moving coil galvanometer is a highly sensitive instrument due to which it can be used to detect the presence of current in any given circuit. If a galvanometer is a connected in a Wheatstone’s bridge circuit, the pointer in the galvanometer shows null deflection, i.e no current flows through the device. The pointer deflects to the left or right depending on the direction of the current.
The galvanometer can be used to measure:
a) the value of current in the circuit by connecting it in parallel to low resistance.
Conversion Of Galvanometer To Ammeter
A galvanometer is converted into an ammeter by connecting it in parallel with a low resistance called shunt resistance. Suitable shunt resistance is chosen depending on the range of the ammeter.
Conversion Of Galvanometer To Voltmeter
A galvanometer is converted into a voltmeter by connecting it in series with high resistance. A suitable high resistance is chosen depending on the range of the voltmeter.
Advantages And Disadvantages Of A Moving Coil Galvanometer
- High sensitivity.
- Not easily affected by stray magnetic fields.
- The torque to weight ratio is high.
- High accuracy and reliability.
- It can be used only to measure direct currents.
- Develops errors due to factors like aging of the instrument, permanent magnets and damage of spring due to mechanical stress.
MCQ on Galvanometer
Q1. What is moving coil galvanometer used for?
a) Measurement of voltage only
b) Measurement of resistance
c) Measurement of small currents
d) Measurement of electric field
Q2. Pick out the expression for galvanometer constant from the following?
a. G = kNAB
b. G = k × NAB
c. G = NABk
Q3. Find the true statement.
a) Ammeter is an instrument used to measure potential difference across any element in a circuit
b) Voltmeter is an instrument used to measure current in a circuit
c) Galvanometer constant is dimensionless
d) Current sensitivity is expressed as the exact reverse of the galvanometer constant
Q4. Voltage sensitivity and current sensitivity are related.
Q5. How is a galvanometer converted into an ammeter?
a) By connecting a high resistance shunt in parallel to the galvanometer
b) By connecting a low resistance shunt in parallel to the galvanometer
c) By connecting a high resistance shunt in series with the galvanometer
d) By connecting a low resistance shunt in series with the galvanometer
Q6. How is galvanometer converted into a voltmeter?
a) By connecting a high resistance multiplier in parallel to the galvanometer
b) By connecting a low resistance multiplier in parallel to the galvanometer
c) By connecting a low resistance multiplier in series with the galvanometer
d) By connecting a high resistance multiplier in series with the galvanometer
Q7. What should be the value of shunt resistance of the ammeter in order to increase its range?
a) S = (n – 1) G
b) S = (n – 1)/G
c) S = G(n–1)
d) S = G(n+1)
Q8. An ideal ammeter has infinite resistance and an ideal voltmeter has zero resistance.
Q9. A galvanometer has resistance of 10 ohms and a full scale deflection is produced by 5 milli amperes. What is the value of resistance that should be connected in series with it in order to enable it to read 2V?
a) 20 Ω
b) 10 Ω
c) 40 Ω
d) 30 Ω
Q10. When 0.010A current flows through a moving coil galvanometer, it gives full deflection. Then, it is converted into a voltmeter which gives a reading of 10 V using an external resistance of 950 Ω. What is the resistance of the galvanometer (G)?
a) 40 Ω
b) 50 Ω
c) 60 Ω
d) 70 Ω
Q11. A moving coil galvanometer has the following characteristics – Number of turns of coil = 50; Area of coil = 70 mm2; Resistance of coil = 30 Ω; Flux density of radial field = 0.1 T; Torsional constant of suspension wire = 7 × 10-8 N m/rad. Calculate the current and voltage sensitivity.
a) 10 div/mA, 0.166 div/mV
b) 15 div/mA, 0.115 div/mV
c) 5 div/mA, 0.167 div/mV
d) 20 div/mA, 0.100 div/mV
Q12. What is the value of the shunt resistance (S) required if a galvanometer has a resistance f 50 Ω and a maximum of 0.05A current that can be passed through it? The ammeter range is changed to 20 A.
a) 0.120 Ω
b) 0.125 Ω
c) 0.130 Ω
d) 0.145 Ω
Q13. The current passing through a galvanometer is 30 mA, resistance of the galvanometer is 50 Ω and a shunt is 1 Ω is connected to the galvanometer. What is the maximum current that can be measured by this ammeter?
a) 1.53 A
b) 15.3 A
c) 0.153 A
d) 153 A